All sonic anemometers need to correct for transducer and structural shadowing in their velocity measurementsAuthor(s): John M. Frank; William J. Massman; Edward Swiatek; Herb A. Zimmerman; Brent E. Ewers
Source: Journal of Atmospheric and Oceanic Technology. 33: 149-167.
Publication Series: Scientific Journal (JRNL)
Station: Rocky Mountain Research Station
Download Publication (1.0 MB)
Related Research Highlights
Ecosystem Fluxes are Underestimated Due to Measurement Tool Errors
Sonic anemometry is fundamental to all eddy-covariance studies of surface energy and ecosystem carbon and water balance. Recent studies have shown that some nonorthogonal anemometers underestimate vertical wind. Here it is hypothesized that this is due to a lack of transducer and structural shadowing correction. This is tested with a replicated intercomparison experiment between orthogonal (K-probe, Applied Technologies, Inc.) and nonorthogonal (A-probe, Applied Technologies, Inc.; and CSAT3 and CSAT3V, Campbell Scientific, Inc.) anemometer designs. For each of the 12 weeks, five randomly selected and located anemometers were mounted both vertically and horizontally. Bayesian analysis was used to test differences between half-hourly anemometer measurements of the standard deviation of wind (su, sy, and sw) and temperature, turbulent kinetic energy (TKE), the ratio between vertical/horizontal TKE (VHTKE), and sensible heat flux (H). Datasets were analyzed with various applications of transducer shadow correction. Using the manufacturer’s current recommendations, orthogonal anemometers partitioned higher VHTKE and measured about 8%-9% higher and ~10% higher H.This difference can be mitigated by adding shadow correction to nonorthogonal anemometers. The horizontal manipulation challenged each anemometer to measure the three dimensions consistently, which allowed for testing two hypotheses explaining the underestimate in vertical wind. While measurements were essentially unchanged when the orthogonal anemometers were mounted sideways, the nonorthogonal anemometers changed substantially and confirmed the lack of shadow correction. Considering the ubiquity of nonorthogonal anemometers, these results are consequential across flux networks and could potentially explain half of the ~20% missing energy that is typical at most flux sites.
- You may send email to email@example.com to request a hard copy of this publication.
- (Please specify exactly which publication you are requesting and your mailing address.)
- We recommend that you also print this page and attach it to the printout of the article, to retain the full citation information.
- This article was written and prepared by U.S. Government employees on official time, and is therefore in the public domain.
Frank, John M.; Massman, William J.; Swiatek, Edward; Zimmerman, Herb A.; Ewers, Brent E. 2016. All sonic anemometers need to correct for transducer and structural shadowing in their velocity measurements. Journal of Atmospheric and Oceanic Technology. 33: 149-167.
Keywordssonic anemometers, carbon balance, water balance, wind
- A Bayesian model to correct underestimated 3-D wind speeds from sonic anemometers increases turbulent components of the surface energy balance
- Underestimates of sensible heat flux due to vertical velocity measurement errors in non-orthogonal sonic anemometers
- Coordinate rotation-amplification in the uncertainty and bias in non-orthogonal sonic anemometer vertical wind speeds
XML: View XML